|NI-SCOPE (English | Japanese)|
Both Configure Vertical and Configure Chan Characteristics affect settings that may be different for each channel. Therefore, if you have a two-channel digitizer, you can call each of these functions twice, changing the channel name parameter between 0 and 1. If you want to set these parameters to the same value for multiple channels, use a channel list such as 0,1.
With the Configure Vertical function, you can adjust the vertical range, vertical offset, vertical coupling, and the probe attenuation parameters.
With the Configure Chan Characteristics function, you can adjust the input impedance and the and the max input frequency parameters.