|NI-DCPower (English | Japanese)|
Short Name: Reset Average Before Measurement
Property of niDCPower
Specifies whether the measurement returned from any measurement call starts with a new measurement call (TRUE) or returns a measurement that has already begun or completed (FALSE).
|Note This property is not supported by all devices. Refer to Supported Properties by Device for information about supported devices.|
When you set the Samples to Average property in the Running state, the output channel measurements might move out of synchronization. While NI-DCPower automatically synchronizes measurements upon the initialization of a session, you can force a synchronization in the running state before you run the niDCPower Measure Multiple VI. To force a synchronization in the running state, set the Reset Average Before Measurement property to TRUE, and then run the niDCPower Measure Multiple VI specifying all channels in the channel name parameter. You can set the Reset Average Before Measurement property to FALSE after the niDCPower Measure Multiple VI completes.
Default Value:Refer to Supported Properties by Device for the default value by device.
The following table lists the characteristics of this property.