|NI-DCPower (English | Japanese)|
ViStatus niDCPower_Abort(ViSession vi);
Transitions the NI-DCPower session from the Running state to the Committed state. If a sequence is running, it is stopped. Any configuration functions called after this function are not applied until the niDCPower_Initiate function is called. If power output is enabled when you call the niDCPower_Abort function, the output channels remain in their current state and continue providing power.
Use the niDCPower_ConfigureOutputEnabled function to disable power output on a per channel basis. Use the niDCPower_reset function to disable output on all channels.
Refer to the Programming States topic in the NI DC Power Supplies and SMUs Help for information about the specific NI-DCPower software states.
|vi||ViSession||Identifies a particular instrument session. vi is obtained from the niDCPower_InitializeWithChannels function.|
|Status||ViStatus||Reports the status of this operation. To obtain a text description of the status code, call the niDCPower_error_message function. To obtain additional information concerning the error condition, call the niDCPower_GetError function.|