|NI-RFSG 18.1 Help|
|NI-RFSG 18.2.1 Help|
|NI-RFSG 19.1 Help|
Configures the NI-RFSG device Reference Clock. The Reference Clock ensures that the NI-RFSG devices are operating from a common timebase. The NI-RFSG device must be in the Configuration state before you call this VI.
Supported Devices: PXIe-5644/5645/5646, PXI/PXIe-5650/5651/5652, PXIe-5653/5654/5654 with PXIe-5696, PXI-5670/5671, PXIe-5672/5673/5673E, PXIe-5820/5840
ref clock source specifies the source of the Reference Clock signal.
ref clock rate specifies the Reference Clock rate, in hertz (Hz). The default value is Auto (-1.0), which allows NI-RFSG to use the default Reference Clock rate for the device or automatically detect the Reference Clock rate, if supported.
error in describes error conditions that occur before this node runs. This input provides standard error in functionality.
error out contains error information. This output provides standard error out functionality.
Refer to the following VIs for examples of using the niRFSG Configure Ref Clock VI.
• RFSG Getting Started Single Tone Generation VI: labview\examples\instr\niRFSG
• RFSG Arbitrary Waveform Generation (Arb) VI: labview\examples\instr\niRFSG
• RFSG Getting Started Script (Script) VI: labview\examples\instr\niRFSG