|NI-RFSG 18.1 Help|
|NI-RFSG 18.2.1 Help|
|NI-RFSG 19.1 Help|
Corrects for any signal drift due to temperature variation when generating the same signal for extended periods of time without a parameter change. Under normal circumstances of short-term signal generation, NI-RFSG performs thermal correction automatically by ensuring stable power levels, and you do not need to call this VI. Use this VI when generating the same signal for a long period of time in a temperature-fluctuating environment. The NI-RFSG device must be in the Generation state before you call this VI.
Supported Devices: PXIe-5611, PXI/PXIe-5650/5651/5652, PXIe-5653/5654/5654 with PXIe-5696, PXI-5670/5671, PXIe-5672/5673/5673E, PXIe-5840
error in describes error conditions that occur before this node runs. This input provides standard error in functionality.
error out contains error information. This output provides standard error out functionality.
Refer to the RFSG Thermal Correction VI in the labview\examples\instr\niRFSG directory for an example of using the niRFSG Perform Thermal Correction VI.