|NI-RFSA 18.1 Help|
|NI-RFSA 18.2.1 Help|
|NI-RFSA 19.1 Help|
ViStatus niRFSA_InitWithOptions(ViRsrc resourceName, ViBoolean IDQuery, ViBoolean reset, ViConstString optionString, ViSession* newVi);
Creates a new session for the device. This function sets the initial value of certain attributes and sends initialization commands to reset all hardware modules to a known state necessary for NI-RFSA operation.
To create a new session, pass the downconverter resource name for the RF vector signal analyzer to the resource name parameter.
You can access the device session this VI creates using the NI-RFSA Soft Front Panel (SFP). Accessing the device session with the SFP can help you debug your code. Refer to Debugging Your Application Using SFP Session Access for more information about accessing your session with the SFP.
Supported Devices: PXI-5600, PXIe-5601/5603/5605/5606 (external digitizer mode), PXIe-5644/5645/5646, PXI-5661, PXIe-5663/5663E/5665/5667/5668, PXIe-5693/5694/5698, PXIe-5820/5840
Driver Setup Options
|resourceName||ViRsrc||Specifies the resource name of the device to initialize.
For NI-RFSA devices, the syntax is the device name specified in MAX. The typical default name for your device in MAX is PXI1Slot2. You can rename your device by right-clicking the name in MAX, selecting Rename from the drop-down menu, and entering a new name. You can also pass in the name of an IVI logical name configured with the IVI Configuration utility. For additional information, refer to the Installed Devices»IVI topic of the Measurement & Automation Explorer Help.
|IDQuery||ViBoolean||Specifies whether NI-RFSA performs an ID query. When you perform an ID query, NI-RFSA verifies the device you initialize is supported.
|reset||ViBoolean||Specifies whether the NI-RFSA device is reset during the initialization procedure.|
|optionString||ViConstString||Sets the initial value of certain attributes for the session. The attributes shown in the following table are used in this parameter.
The format of this string is "AttributeName=Value", where AttributeName is the name of the attribute and Value is the value to which the attribute will be set. For example, you can simulate the PXIe-5663 using the following strings:
"Simulate=1, DriverSetup=Model:5601; Digitizer:5622; LO:5652; LOBoardType:PXIe".
To set multiple attributes, separate their assignments with a comma.
Refer to Driver Setup Options for more information about the driver setup string.
|vi||ViSession||Identifies your instrument session.|
|status||ViStatus||Returns the status code of this operation. The status code either indicates success or describes an error or warning condition. Examine the status code from each call to an NI-RFSA function to determine if an error has occurred.
To obtain a text description of the status code and additional information about the error condition, call the niRFSA_GetError function. To clear the error information from NI-RFSA, call the niRFSA_ClearError function.
The general meaning of the status code is as follows: