|LabVIEW 2018 Sound and Vibration Toolkit Help|
|LabVIEW 2019 Sound and Vibration Toolkit Help|
Owning Palette: Configure Advanced Settings VIs
Requires: Sound and Vibration Toolkit
Sets the propagation time, expressed in number of samples, introduced by the data acquisition board used to perform the swept-sine measurement and the propagation time, expressed in milliseconds, introduced by the device under test (DUT).
|swept sine task in specifies the swept-sine parameters. |
|device propagation delay [samples] specifies the propagation delay, in number of samples, for the device used to perform the measurement. The default is 1 sample.|
|DUT propagation time [ms] specifies the propagation time for the device under test (DUT) in milliseconds. The default is 0 ms.|
|error in describes error conditions that occur before this node runs. This input provides standard error in functionality.|
|swept sine task out returns the swept-sine parameters.|
|error out contains error information. This output provides standard error out functionality.|
Refer to the Frequency Response with Swept Sine (DAQmx) VI in the labview\examples\Sound and Vibration\Swept Sine directory for an example of using the SVT Set Swept Sine Propagation Delays VI.