Download Help (Windows Only) 
Owning Palette: Jitter VIs
Requires: Jitter Analysis Toolkit
Measures the duty cycle distortion in a waveform, or the amount by which the mean positive width of cycles differs from the mean negative width.
Duty cycle distortion is a type of deterministic jitter in which the clock generates positive pulses that are not equal to negative pulses.
level crossings contains information about the locations of level crossings in the original waveform. You can use the Level Crossing VI to generate this cluster.
 
error in describes error conditions that occur before this node runs. This input provides standard error in functionality.  
slope specifies the direction of the transition at which to begin the measurement. This VI begins measuring at the first transition of that direction in the waveform.


DCD (s) returns the amount by which the mean pulse duration of positive pulses differs from the mean pulse duration of negative pulses.  
DCD (%) is the duty cycle distortion as a percentage of the mean cycle period. This VI computes the DCD (%) as (1/mean period × DCD (s) × 100).  
error out contains error information. This output provides standard error out functionality. 