|Download Help (Windows Only)|
Location: In the navigation bar at the top of the page, click the arrow next to the Configuration button and select Devices in the pull-down menu. Double-click a device in the list on the Devices tab and click the Channels tab.
This tab defines the properties of the physical channel on the device. The type of data the channel acquires and the types of measurements it performs are defined by the asset to which it is mapped.
|Configure multiple channels at once.||<Ctrl>- or <Shift>-click the channels in the table at the left. The tabs in the right-hand pane update to show only properties that the selected channels share. Use the tabs to configure all selected channels at once.|
|Learn about the sample rates at which channels acquire data.||The sample rate you set on the Device Configuration page»Device Properties tab applies to channels on NI 9232 modules. All static C Series modules have a sample rate of 1 Hz. For more information, refer to Frequently Asked Questions about Sample Rates and Durations.|