|Download Help (Windows Only)|
This topic contains information for calibrating National Instruments PXIe-4357 SC Express modules using Calibration Executive.
Calibration Executive Procedure Features:
||Selectable Test Points
Approximate Test Time:
|Verify & Adjust
|Test Equipment|||||Test Conditions|||||Device Setup|||||Test Limit Equations|
The following table lists the test equipment required for calibrating PXIe-4357 modules.
|Calibrator||Fluke 5520A||Use a calibrator that can provide resistance values in the range of 0 Ω to 400 Ω with 0.01 Ω resolution, an accuracy of 90 ppm or better, and automatic lead-wire compensation.|
|PXI Express Chassis||PXIe-1062Q||—|
|Connection Accessory||TB-4357 terminal block||Screw terminal accessory for the PXIe-4357.|
The following setup and environmental conditions are required to ensure the PXIe-4357 meets published specifications.
The calibration procedure for the PXIe-4357 requires that you perform an Ohms Zero operation on the calibrator during setup.
The following test limits are derived from the published specifications.
TestLimits = TestValue ± (TestValue * Gain Error + Offset Error)