|Download Help (Windows Only)|
This topic contains information for calibrating the National Instruments 9361 using Calibration Executive.
Calibration Executive Procedure Features:
||Selectable Test Points
Approximate Test Time:
|Test Equipment|||||Test Conditions|||||Device Setup|||||Test Limit Equations|
The following table lists the test equipment required for calibrating NI 9361 modules.
|Frequency Standard||Keysight 33250A||A highly stable clock source that is able to generate 1 MHz, 5 V TTL output with ≤ ±5 ppm total accuracy|
|Connection Accessory||NI 9923||—|
The following setup and environmental conditions are required to ensure the NI 9361 meets published specifications.
The following test limits are derived from the published specifications found in the NI 9361 Calibration Procedure.
Accuracy = ± 50.0 ppm