|Download Help (Windows Only)|
This topic contains information for calibrating the National Instruments TB-4302C terminal block using Calibration Executive.
Calibration Executive Procedure Features:
||Selectable Test Points
Approximate Test Time: 7 minutes
|Test Equipment|||||Test Conditions|||||Device Setup|||||Test Limit Equations|
The following table lists the test instruments required for calibrating the TB-4302C.
|DMM||PXI-4071||Use a DMM that has an accuracy of 136 ppm or better when measuring 5 Ω in 4-wire mode.|
|4-wire Kelvin probe||Keithley 5808||Use a probe capable of measuring Kelvin resistance.|
The following setup and environmental conditions are required to ensure the TB-4302C meets published specifications.
The following test limits are derived from the published specifications.
TestLimit = TestValue ± [TestValue * (GainError / 100)]