Specifies which instrument function controls the pin(s). By default, all pins are set to Disconnect.
NationalInstruments.ModularInstruments.NIDigital.Fx40 (in NationalInstruments.ModularInstruments.NIDigital.Fx40.dll) Version: 220.127.116.11152 (18.104.22.168152)
|Visual Basic (Declaration)|
Public Enumeration SelectedFunction
public enum SelectedFunction
The pattern sequencer controls the specified pin(s). If a pattern is currently bursting, the pin immediately switches to bursting the pattern. This option disconnects the PPMU.
The I/O switch connecting the instrument channel is open to the I/O connector. If the PPMU is sourcing, it is stopped prior to opening the I/O switch.
Puts the digital driver in a non-drive state, disables the active load, disconnects the PPMU, and closes the I/O switch connecting the instrument channel.
The PPMU controls the specified pin(s) and connects the PPMU. The pin driver is in a non-drive state, and the active load is disabled. The PPMU does not start sourcing or measuring until
or DigitalPpmu.Measure is called.