|Download Help (Windows Only)|
Aborts the scan in progress. Use the niSwitch Initiate Scan VI to initiate an aborted scan.
If the switch module is not scanning, a No Scan In Progress error is returned.
If error in describes an error that had occurred before calling this VI, the VI still attempts to abort the scan. However, if the attempt fails, the VI returns the error information that was passed in from the error in parameter.
|Note If you are using a legacy resource descriptor (SCXI:: or PXI::) and you abort a scan, the switch module returns to a disconnect all state (equivalent to calling the niSwitch Disconnect All Channels VI). If you are using an NI-DAQmx resource descriptor (DAQmx device name), the switch module returns to the state it was in before the scan was initiated.|
error in describes error conditions that occur before this node runs. With the following exception, this input provides standard error in functionality. This node runs normally even if an error occurred previously. If an error occurred before this node runs, the node passes the error in value to error out, and no errors that occur while this node runs are recorded. If an error occurs while this node runs and no error occurred previously, the node sets its own error status in error out. The default is no error.
instrument handle out identifies a particular NI-SWITCH session established with the niSwitch Initialize With Topology VI, the niSwitch Initialize With Options VI, or the niSwitch Initialize VI and used for all subsequent NI-SWITCH calls.
error out contains error information. This output provides standard error out functionality.