|LabVIEW 2018 ELVIS III Toolkit SP2 Help|
|LabVIEW 2019 ELVIS III Toolkit Help|
Owning Palette: Instruments VIs
Requires: ELVIS III Toolkit. This topic might not match its corresponding palette in LabVIEW depending on your operating system, licensed product(s), and target.
Use the Bode Analyzer VIs to configure and control the Bode Analyzer instrument.
Always connect the fgen/1 channel to the scope/1 channel when you use the Bode Analyzer VIs. The instrument uses fgen/1 to output the stimulus signal, uses scope/1 to measure the stimulus signal, and uses scope/2 to scope/4 to measure the response signals. scope/1 serves as the reference channel.
|Bode Calculate Logarithm Sweep Array||Calculates the frequency points of the stimulus signal during a Bode plot sweep.|
|Bode Close||Stops the session and deallocates any resources acquired during the session.|
|Bode Configure Peak Amplitude||Sets the peak amplitude of the FGen/Arb instrument's output signal.|
|Bode Initialize||Establishes communication with the device. Call this VI once per session.|
|Bode Query Peak Amplitude||Checks the peak amplitude of the stimulus signal that the FGen/Arb instrument generates.|
|Bode Read Gain and Phase||Measures the gain and phase difference of the response signals during a plot frequency sweep.|