|NI-SCOPE (English | Japanese)|
|Access||Applies to||Coercion||High Level Functions|
Specifies the position of the Reference Event in the waveform record. When the digitizer detects a trigger, it waits the length of time the NISCOPE_ATTR_TRIGGER_DELAY_TIME attribute specifies. The event that occurs when the delay time elapses is the Reference Event. The Reference Event is relative to the start of the record and is a percentage of the record length. For example, the value 50.0 corresponds to the center of the waveform record and 0.0 corresponds to the first element in the waveform record.
Valid Values: 0.0 – 100.0