|NI-SCOPE (English | Japanese)|
Short Name: Trigger Window High Level
Property of niScope
Pass the upper voltage threshold you want the digitizer to use for window triggering.
The digitizer triggers when the trigger signal enters or leaves the window you specify with the Trigger Window Low Level property and this property.
The values of the Vertical Range property and the Vertical Offset property determine the valid range for the Trigger Window Low Level property on the channel you specify with the Trigger Source property.
The value you pass for this parameter must meet the following conditions:
High Trigger Level <= Vertical Range/2 + Vertical Offset
High Trigger Level >= (-Vertical Range/2) + Vertical Offset
High Trigger Level > Low Trigger Level
The following table lists the characteristics of this property.
|High-level VIs||niScope Configure Trigger Window (Analog Window Ref Trigger)|