|Download Help (Windows Only)|
Short Name: LEInBandEmission Auto Sweep Time Enabled
Property of niBTAnalysis
Specifies whether to enable auto sweep time for LEInBandEmission measurements.
When you set this property to True, the toolkit sets the value for the LEInBandEmission Sweep Time property based on the value of the LEInBandEmission Measurement Method property. If you set the LEInBandEmission Measurement Method property to Sweep, the toolkit sets the sweep time to 5 ms. If you set the LEInBandEmission Measurement Method property to FFT, the toolkit sets the sweep time to 10 ms. The toolkit uses a higher sweep time for the FFT method compared to Sweep method because the variations in channel power measurements are higher for lower sweep times in the FFT method.
If you set the LEInBandEmission Auto Sweep Time Enabled property to False, the toolkit uses the sweep time, which you set in the LEInBandEmission Sweep Time property, to analyze the signal.
The default value is True.
Disables auto sweep time.
Enables auto sweep time.
The following table lists the characteristics of this property.
|High-level VIs||niBT SA LEInBandEmission Set Auto Sweep Time Enabled|