|Download Help (Windows Only)|
Parent Topic: Available Viewers
Supported Sensors: Vibration sensors that acquire waveform data, such as acceleration. Assign the sensors to a data group with a tachometer as the speed reference.
The Order Waveform viewer displays the waveform data resampled to a constant number of samples per revolution. Use this viewer with the Order Spectrum viewer to see data when the equipment speed is changing or when comparing data acquired at different machine speeds.
|Show baselines.||Click Show Baselines on the viewer toolbar.|
|Display the 1x waveform or a filtered 1x waveform.||This viewer can display four representations of a waveform: waveform, auto-correlated waveform, 1x waveform, or filtered 1x waveform. To configure which representations are displayed, click Waveform Type and use the four radio buttons in the resulting dialog box. If you choose to show the filtered 1x waveform, you can also configure the lower and upper bound values in this dialog box.
|Apply single or double integration or differentiation to data.||
Click Integration on the viewer toolbar and select an option from the resulting dialog box.
Related link: Integrating and Differentiating Data
|Display only a certain number of revolutions.||If a measurement is from a sensor with a tachometer reference, you can display only data acquired during a specified number of revolutions. Click Settings on the viewer toolbar, select the Show fixed number of revolutions checkbox, and enter the number you want in the # revolutions textbox. The viewer displays the first n revolutions in the measurement. If this setting is disabled, the viewer displays the entire waveform.|
|Mark the locations of tachometer pulses.||If a measurement is from a sensor with a tachometer reference, you can display bold marks on the curve at each position where a tachometer pulse occurred. Click Settings on the viewer toolbar, and select the Show tachometer reference checkbox.|
|Enable AC coupling of waveforms.||To remove the DC offset from waveform data, click Settings on the viewer toolbar, and select the AC-couple waveform data checkbox.|
|Compare measurements from different sources or times.||
Create a data event reference.
Related link: Comparing Measurements from Different Sources or Times
|Select the source sensor.||If you have more than one feature selected in the asset tree, you can select the source sensor for the viewer by clicking Choose Assets on the viewer toolbar.|