|Download Help (Windows Only)|
This topic contains information for calibrating the NI PXIe-4339 using Calibration Executive.
Calibration Executive Procedure Features:
||Selectable Test Points
Approximate Test Time:
|Verify & Adjust
|Test Equipment|||||Test Conditions|||||Device Setup|||||Test Limit Equations|
The following table lists the test equipment required to calibrate the PXIe-4339.
|Calibrator||Fluke 5700||If this instrument is unavailable, use a calibrator that can provide DC voltage values in the range of 0 to ±10 V with an accuracy of 50 ppm of output or better and a DC offset error of 4 µV or better.|
|PXI Express Chassis||PXIe-1062Q||—|
|* The terminal block variants TB-4339, TB-4339B, and TB-4339C differ only in resistance; the measurements are the same regardless of the variant used.|
The following setup and environmental conditions are required to ensure the PXIe-4339 meets published specifications.
The following test limits are derived from the published specifications.
Refer to the NI PXIe-4339 Device Specifications for Gain Error and Offset Error values.