|Download Help (Windows Only)|
This topic contains information for calibrating the NI 9260 using Calibration Executive.
Calibration Executive Procedure Features:
|Adjustment||Manual Mode||Selectable Test Points
Approximate Test Time: 13 minutes
|Test Equipment|||||Test Conditions|||||Device Setup|||||Test Limit Equations|
The following table lists the test equipment required for calibrating NI 9260.
|DMM||PXI-4070||If this instrument is unavailable, use a multiranging 6 1/2-digit DMM with an accuracy of at least ±40 ppm.|
The following setup and environmental conditions are required to ensure the NI 9260 meets published specifications.
|Note If the NI module fails after calibration, return it to NI for repair or replacement.|
The following test limits are derived from the published specifications found in the NI 9260 Calibration Procedure.
TestLimit = TestValue ± (TestValue * (GainError / 100) + (FS * (OffsetError / 100)))