|RFmx Demod 19.1 Help|
|RFmx Demod 20.0 Help|
|RFmx Demod 20.6 Help|
Use the VIs on this palette to configure various parameters for RFmxDemod measurements.
|RFmxDemod Configure RF||
Configures the RF properties of the signal by specifying the selector string.
|RFmxDemod Configure Frequency||
Configures the expected carrier frequency of the RF signal that needs to be acquired. The signal analyzer tunes to this frequency.
|RFmxDemod Configure Reference Level||
Configures the reference level that represents the maximum expected power of an RF input signal.
|RFmxDemod Auto Level||
Examines the input signal to calculate the peak power level and sets it as the value of the Reference Level property. Use this VI to help calculate an approximate setting for the reference level.
|RFmxDemod Configure External Attenuation||
Configures the attenuation of a switch (or cable) connected to the RF IN connector of the signal analyzer.
|RFmxDemod Configure Trigger||
Configures the Reference Trigger to use to acquire the signal.
|RFmxDemod Send Software Edge Trigger||
Sends a trigger to the device when you use the RFmxDemod Configure Trigger VI to choose a software version of a trigger, and the device is waiting for the trigger to be sent. You can also use this VI to override a hardware trigger.
Use the VIs on your palette to configure analog demodulated measurements.
Use the VIs on this palette to configure digital demodulated measurements.