|Download Help (Windows Only)|
The following figure represents the NI PXI-2510 in the independent topology.
signal connected to DUT1 is routed to busA
|Note The pass-through channels (chN to dutN) on the PXI-2510 are connected in its power on or reset state. The niSwitch Reset and niSwitch Disconnect All Channels VIs or niSwitch_reset and niSwitch_DisconnectAll functions will connect the pass-through channels (chN to dutN).
Special attention needs to be given to the behavior of the relays associated with the pass-through channels (k0...k67). These relays (k0...k67) are in their open state when the pass-through channels (chN to dutN) are connected and vice versa. If you intend to use the niSwitch Relay Control VI or niSwitch_RelayControl function to operate these relays, please note that closing relays (k0...k67) will disconnect the pass through channels (chN to dutN).
Because the pass-through channels (chN to dutN) are connected in its power-on or reset state, these channels may need to be disconnected during a scanning operation. In order to disconnect the channels during a scan, set the Scan Mode property to None, or the NISWITCH_ATTR_SCAN_MODE attribute to NISWITCH_VAL_NONE.
The following figure identifies the pins for the NI PXI-2510.